UMP3-1UltraMicroPump (UMP3) (one) with SYS-Micro4 Controller
- Related Products
There are 1 images available to view - click to enlarge and scroll through the product gallery.
/ Download as PDF
/ Download as PDF
- Versatile injector uses microsyringes to deliver picoliter volumes
- Syringes are easily installed, just snap the barrel into the clamps
- Threaded shaft drives syringe plunger allowing smooth and accurate movement
With its digital controller, UltraMicroPump III can dispense as little as 600 picoliters per incremental advance of the syringe piston (using a 0.5μL syringe). Syringes may be filled externally and then inserted into the pump or filled while mounted in the pump. Fluids injected or withdrawn are held entirely within the micro syringe to maintain a low fluid dead volume.
For positioning, the UltraMicroPump III may be attached to any of several WPI micropositioners such as the M3301 (manual), DC3001 (motorized), or any manual stereotaxic manipulator.
An Integral component in the UMPIII system is a microprocessor-based controller, SYS-MICRO4, which provides an "intelligent" and easy-to-use interface to up to four syringe pumps. Operating parameters are set with the membrane keypad and LCD display. From the keypad the user can select the following functions: set pump to infusion or withdrawal mode, enter the volume to be infused or withdrawn, rate of delivery, and syringe type as well as synchronize the starting and stopping of any combination of syringe pumps. User parameters can be stored in the device's "non-volatile" memory for instant recall when the unit is powered on. An optional footswitch can be plugged into a connector on the rear of the controller for "hands free" start/stop operation. Computer Control - An RS-232 port on the rear of the controller can be used to connect it to a computer for use with computer control programs.
|Total # of Steps||20,000(63mm travel)|
|Minimum Dispensing Volume||0.58nL/step (10μL syringe)|
|Linear Motion per Step||3.175μm|
|Weight||325g (11.5 oz.)|
|Mounting Rod Diameter||7.9mm (0.31 in.)|
|Controller Power Requirements||2A, 12VDC|
|Dimensions||Ø32mm x 190mm (1.3 in. x 7.5 in.)|
Precision is increased eight fold.
UltraMicroPump (head only)
Castellano, O., Arji, M., & Sancho, C. (2013). Chronic administration of risperidone in a rat model of schizophrenia: A behavioural, morphological and molecular study. Behavioural brain …. Retrieved from http://www.sciencedirect.com/science/article/pii/S0166432812008261
Donnelley, M., & Morgan, K. (2013). Variability of In Vivo Fluid Dose Distribution in Mouse Airways Is Visualized by High-Speed Synchrotron X-Ray Imaging. Journal of aerosol …. Retrieved from http://online.liebertpub.com/doi/abs/10.1089/jamp.2012.1007
Gosselin, D., Bellavance, M., & Rivest, S. (2013). IL-1RAcPb signaling regulates adaptive mechanisms in neurons that promote their long-term survival following excitotoxic insults. Frontiers in cellular …. Retrieved from http://www.ncbi.nlm.nih.gov/pmc/articles/PMC3573345/
Lloyd-Burton, S., & York, E. (2013). SPARC Regulates Microgliosis and Functional Recovery following Cortical Ischemia. The Journal of …. Retrieved from http://www.jneurosci.org/content/33/10/4468.short
Lu, X., Huxley, V., & Kassab, G. (2013). Endothelial barrier dysfunction in diabetic conduit arteries: a novel method to quantify filtration. American Journal of …. Retrieved from http://ajpheart.physiology.org/content/304/3/H398.short
Ma, Z., Gulia-Nuss, M., Zhang, X., & Brown, M. (2013). Effects of the Botanical Insecticide, Toosendanin, on Blood Digestion and Egg Production by Female Aedes aegypti (Diptera: Culicidae): Topical Application and. Journal of medical entomology. Retrieved from http://www.bioone.org/doi/pdf/10.1603/ME12119
Montgomery, S., & Narrow, W. (2013). Chronic Neuron-and Age-Selective Down-Regulation of TNF Receptor Expression in Triple-Transgenic Alzheimer’s Disease Mice Leads to Significant Modulation of. The American journal of …. Retrieved from http://www.sciencedirect.com/science/article/pii/S0002944013002095
Rangel, A., & Race, B. (2013). Non-amyloid and amyloid prion protein deposits in prion-infected mice differ in blockage of interstitial brain fluid. Neuropathology and …. Retrieved from http://onlinelibrary.wiley.com/doi/10.1111/j.1365-2990.2012.01303.x/full
Silberman, Y. (2013). A Corticotropin Releasing Factor Pathway for Ethanol Regulation of the Ventral Tegmental Area in the Bed Nucleus of the Stria Terminalis. The Journal of …. Retrieved from http://www.jneurosci.org/content/33/3/950.short
Wen-Ming, M., & Hua-Wei, L. (2013). Chitin-binding proteins of Artemia diapause cysts participate in formation of the embryonic cuticle layer of cyst shells. Biochemical …. Retrieved from http://www.biochemj.org/bj/449/bj4490285.htm